The inner water pipe is made from antiseptic material, clinical fluids, such as hydrogen peroxide, chlorhexidine and sodium Hypochlorite, etc, can be used perfectly under the mode of auto-water supply system so that the performance of perio and endo is improved dramatically. The handpiece is detachable and can be autoclaved to the temperature of 135? And the pressure of 0.22MPa. Automatic frequency tracking ensures that the machine always works on the best frequency and performs more steadily. Controlled by PC, easy operation and more efficient for scaling.
These featurs make SK-D1 become a new generation product is the world dental market. Pay attention to its Three Highlight Features! 1 It uses voltage transformer rather than AC adapter to avoid the feeling of pain when scaling!2 The main unit uses fireproof material. 3 The auto-water supply system adapts diaphragm pump, which will be more durable than peristaltic pump. Please leave a message about which type you need, 110v or 220v. Main unit input:24V1.3A. Output primary tip vibration excursion:=100m. Output tip vibration frequency:28kHz±35Hz. Output power:3W to 20W.
Main unit fuse:T1.6AL 250V. Adapter fuse:T0.5AL 250V. Weight of main unit:1.17g. Weight of adapter:1.2kg. Type of protection against electric shock:Class II equipment.
Degree of protection against electric shock:Type BF applied part. Degree of protection against harmful ingress of water:Ordinary equipment. Protection degree against water(used on the foot switch):IPX1. Applied part of the equipment:handpiece and tip.Degree of safety of application in the presence of a Flammable Anaesthetic Mixture with air or with Oxygen or Nitrous Oxide. Equipment not suitable for being used in the presence of a flammable anaesthetic mixture with air or with oxygen or nitrous oxide. 5Scaling tip GD1 GD2 GD3 GD4 PD1. The sale of this item may be subject to regulation by the U. Food and Drug Administration and state and local regulatory agencies. IMPORTANT INFORMATION ON OUR POLICY AND FEEDBACK. Feedback if you are satisfied with our items. We try to be fair and accurate with all of our product listings and descriptions. If there is a problem, we can work together to resolve it. We strive to always be extremely fair and always try to work with reasonable customers. We will always make every attempt possible to resolve issues if we made a mistake. Please do not assume a mistake is intentional.
Let us know if our service could be better! Tracking info of your parcel is available. The following form is for your reference.
Please repack the item carefully. Il 29-dic-16 alle 06:39:23 CET, il venditore ha aggiunto le seguenti informazioni sull'oggetto. Il 29-dic-16 alle 06:40:17 CET, il venditore ha aggiunto le seguenti informazioni sull'oggetto. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Handpieces & Instruments\Ultrasonic Scalers".The seller is "dentalseasky2011" and is located in this country: US. This item can be shipped worldwide.